E3S Web Conf.
Volume 49, 2018SOLINA 2018 - VII Conference SOLINA Sustainable Development: Architecture - Building Construction - Environmental Engineering and Protection Innovative Energy-Efficient Technologies - Utilization of Renewable Energy Sources
|Number of page(s)||8|
|Published online||13 August 2018|
QuickSun 830A module solar simulator. Study of mini PV modules
Faculty of Mathematics and Natural Sciences, University of Rzeszow, 35-959 Rzeszow, Poland
* Corresponding author: email@example.com
QS830A module solar simulator is applied to measure electrical parameters of standard PV modules based on mono/polycrystalline silicon solar cells, large format photovoltaic (PV) modules 150 cm x 220 cm, with effective measurements time of 2 ms / 4 ms, flush pulse duration 3 ms / 10 ms (one flash tube / two flash tubes), and non-uniformity less than 2%. In order to comply with the Class AAA tolerances of the standard IEC 60904-9 Edition 2.0 of QuickSun 830A simulators (Endeas Oy, Finland), proprietary optical system behind the light source filament has been developed for filtering spectrum and improving irradiance nonuniformity simultaneously. Non-uniformity parameter for 14% of total modules testing area, i.e. 0.65 m2, will be appointed in this work.
© The Authors, published by EDP Sciences, 2018
This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.