E3S Web Conf.
Volume 78, 20192018 International Seminar on Food Safety and Environmental Engineering (FSEE 2018)
|Number of page(s)||5|
|Section||Environmental Biotechnology and Environmental Equipment|
|Published online||15 January 2019|
Gold-nanoparticles-graphene modified glassy carbon electrode for trace detection of lead ions
College of Life Information Science and Instrument Engineering, Hangzhou Dianzi University, Hangzhou 310018, China
2 Department of Chemistry, Chungnam National University, Daejeon, Korea
* Corresponding author: email@example.com,cn
In order to fulfill rapid and trace detection of heavy metals, in the study, gold-nanoparticlesgraphene modified glass carbon electrodes were used to detect lead(Pb(II)) ions. The lead ion (Pb(II)) standard sample in acetic acid-sodium acetate buffer solution was detected by the modified electrode, and the detection conditions were optimized. Under the optimized conditions, in the buffer solution with pH=4.5, the lead ions peak current showed excellent linear relationship with the lead ion concentration in the range of 0.2~50μg/L (R2=0.995), and the minimum detection limit was 0.34μg/L. At the same time, the actual water samples were determined by nano-gold-graphene/glass carbon electrodes for lead ions, and the recovery ranges was 92.94%~101.47%, which showed that the gold-nanoparticles-graphene/glass carbon electrodes possessed significant detection performance for the actual water samples.
© The Authors, published by EDP Sciences, 2019
This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.