E3S Web of Conferences
Volume 1, 2013Proceedings of the 16th International Conference on Heavy Metals in the Environment
|Number of page(s)||4|
|Section||Analytical Chemistry II|
|Published online||23 April 2013|
Direct Determination of Zinc, Cadmium, Lead, Copper Metal in Tap Water of Delhi (India) by Anodic Stripping Voltammetry Technique
Department of Forensic Medicine and Toxicology, All India Institute of Medical Sciences (AIIMS), New Delhi, India - 110029
a firstname.lastname@example.org, +91-99113781
Salts of Zinc, Cadmium, Lead and Copper are taken incidentally or accidently and has become of great toxicological importance having toxic effect. In the present study direct determination of Zn, Cd, Pb and Cu metal was carried out from tap water of Delhi (India) using differential pulse anodic stripping Voltammeter (DPASV) at Hanging mercury dropping electrode (HMDE).Determination of Zn, Cd, Pb, Cu was done using Ammonium acetate buffer (pH 4.6) with a sweep rate (scan rate) of 59.5 mV/s and pulse amplitude 50mV by HMDE by standard addition method. The solution was stirred during pre-electrolysis at -1150mV (vs. Ag/AgCl) for 90 seconds and the potential was scanned from -1150V to +100V (vs..Ag/AgCl). As a result the minimum level of Zn, Cd, Pb, Cu was Zero and the concentration observed in the tap water sample of Delhi (India) was determined as 0.174 mg/L-1, 0.001 mg/L-1, 0.002 mg/L-1, 0.011 mg/L-1 respectively.
Key words: Anodic stripping Voltammetry / Hanging mercury dropping electrode / Heavy Metals / Zinc / Copper / Cadmium / Lead / Tap water
© Owned by the authors, published by EDP Sciences, 2013
This is an Open Access article distributed under the terms of the Creative Commons Attribution License 2.0, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.