E3S Web Conf.
Volume 118, 20192019 4th International Conference on Advances in Energy and Environment Research (ICAEER 2019)
|Number of page(s)||5|
|Section||Energy Equipment and Application|
|Published online||04 October 2019|
The Influence of Harmonics on Error Stability of Energy Meter Verification Device and Its Suppression Measures
Electric Power Research Institute of State Grid Zhejiang Province Electric Power Company, Hangzhou, 310014, China.
2 State Grid Hangzhou Power Supply Company
email@example.com, firstname.lastname@example.org, email@example.com, firstname.lastname@example.org
.In view of the problem of poor error stability of the three-phase energy meter verification device under low current load, the working principle and error calculation method of the device are analyzed. The harmonics content and error data of the device under low current load before and after the installation of the isolation transformer are tested on site. And the role of the isolation transformer in suppressing the harmonics of the device is analyzed. From the harmonics content and error data of the output current of the device before and after the installation of the isolation transformer, it can be seen that the isolation transformer can effectively suppress the harmonics content of the device at the time of low current output and also improve the error stability under low current load.
© The Authors, published by EDP Sciences, 2019
This is an Open Access article distributed under the terms of the Creative Commons Attribution License 4.0, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.